Views:0 Author:Site Editor Publish Time: 2018-04-28 Origin:Site
3. Test operation.
When the multipurpose electronic measuring instrument is adjusted to the appropriate resistance file, the red meter pen and the black meter pen are divided close to the two poles of the capacitance adapter of the measured safety gauge, and the marking value will slowly increase from "000" until the overflow mark "1" is displayed. If "000" is always displayed, it indicates an internal short circuit of the gauge capacitor; if the overflow is always displayed, the internal pole of the gauge capacitor is probably open, or the resistor file chosen is not suitable. When detecting the capacitor period, we should pay attention to the positive electrode of the capacitor with positive charge, and the negative electrode of the capacitor connected with the black meter pen.
The criterion of selecting the resistance file of multimeter is: when the capacitance is small, the high resistance should be used, and the low resistance should be used when the capacitance is large. If large capacity capacitors are estimated with high resistance, the test period will last for a long time because the increase in power is slow.